Glove Box Daily Servicing Manual
1) Fill in the use history in time right after functioning the glove box, which include "user, water oxygen price, what to do"
2) When changing gas cylinders in the glove box, the strain is set to the default worth ± 3mbar
3) The daily operating stress is set in accordance to the working convenience. When not in use for a extended time, the force is set to 1-6mbar
4) On a regular basis test the have on of the gloves, look at the oil level of the vacuum pump (involving the upstream and downstream markings), and add the pump oil properly
five) Clear the drinking water probe at the time each individual two to three months to be certain correct detection. (Also will need to refer to the actual circumstance of the setting in distinct consumer boxes
6) If you frequently use natural solvents in the glove box, it is advisable that you regularly switch the natural and organic solvent adsorbent substance, not later than one calendar year at the newest
seven) Swap the filter element regularly, no afterwards than two several years
8) In accordance to the actual use of your glove box, decide to switch the vacuum pump oil cycle, this kind of as black and yellow pump oil, drinking water or additional natural solvents in the pump oil. The oil alternative cycle should really not exceed 2 several years at the hottest
nine) If a big quantity of organic and natural solvent is inevitably uncovered during the experiment, the cycle requirements to be stopped for the experiment. Just after the experiment, initially clean up and then begin the cycle
ten) When the natural environment in the box is toxic or destructive gas, the exhaust port of the vacuum pump, the exhaust port of the regeneration fuel and innovative technology glovebox the exhaust port of the cleansing valve really should be fatigued to the outdoor with an exhaust pipe (when exhaust is not allowed to be outdoor, the exhaust gas remedy should be done very first).